Simulate signals wired to the DAQ assistant for USB-6009 device
I'm trying to send a signal to the DAQ Assistant Express VI. I watched the movie "Generating a Signal" on the Web site of NOR (www.ni.com/academic/students/learnlabview/generate.htm) and I have my Signal simulate connected directly on the DAQ Assistant, as shown in this film. In my case, the DAQ Assistant sends the signal to a device USB-6009.
However, I received this message:
Error-200077 occurred to the DAQ Assistant
Requested value is not supported for this property value. The value of the property may be invalid because it is in conflict with another property.
asked the value: Sample clock
You select: On-demand
If I select 'On Demand' in my DAQ assistant and run the vi everything works beautifully. However, I need my DAQ assistant to be configured to generate a waveform AC continuous, not output a single alternating current rippling.
What happens here? I did not have this problem before on other devices of NOR. I am using LABView 2010.
Tags: NI Hardware
I have a VI fucntioning, where signal comes 2 simulate VI signals and is being recorded.
I replaced that with the DAQ assistant. I want to know if the wiring is correct and if it will give me the same result as the simulated VI.
Thanks for the reply, yes I did that as well, another way to do it is the 'split signal VI' that automatically separates the signal into 2 channels, most of the high sons being Ch 0
This should be a pretty simple question, but I can't seem to find the answer online and currently do not have the functionality to test this:
I'm using LabVIEW 8.5 and have a VI that imports data from sensor through the DAQ Assistant. In the configuration tab, there is a range of signal input. What happens if my sensor exceeds this range? I get a warning? The default value is the maximum (or minimum)? I was interested in writing a code to display an error that I approach the limits of this range, but did not know if I also need to include code to display an error if the scope is exceeded as well.
Thanks for the help,
The behavior depends on the selected range and the device you are using.
If you are using a device with a single input range is valid, we will use this range, even if you set a smaller minimum and maximum in the DAQ Assistant. So, if your device only supports ±10V and you set the range to ±8V, you will still continue to get valid data after your top sensor 8V until what you approach 10V. When you reach the limit of the extent of your device, the output will be 'rail', and simply return the maximum value until the signal is less than the maximum value again.
Note: A device that is nominally ±10V usually has a go-around (such as ±10.2V) which are usually specced in the manual.
However, if you use a device with several ranges of entry then things become more complex.
NOR-DAQmx player will choose the smallest range that entirely covers the interval you choose. For example, suppose that your device supports the following input range: ±0.2V, ±1, ±5V, ±10V and you choose 0V - 3V as the range in the DAQ assistant. The NOR-DAQmx driver will focus on the input range and the list of the entry lines that your hardware supports and choose the smallest encompassing the entire range that you set. This would be the ±5V, because this is the only beach that contains up to 3V. Thus, all between ±5V input signal is returned and none outside this range will be 'rail' to the maximum or minimum value.
We do this because using small beaches make more efficient use of the resolution of the ADC. So, we try to use the most effective range based on what you ask without picking up a range that will make you miss data.
Let me know if I can clarify it more.
I use the DAQ assistant for analog voltage of an input OR data acquisition card. What is the difference between the rate and the number of samples in the DAQ assistant and what are the units of the two?
The number of samples is how many discrete to measures. Rate (per second) is how fast to acquire the specified number of samples.
If number of samples is 100 and the rate is 1000 samples per second, then the acquisition would take 0.1 second (100 / 1000).
I create an executable file to run on a computer that does not have LabVIEW. The problem is that I am using the DAQ Assistant. The hierarchy of the VI is big enough, and I do not know how to include all the required files to eliminate the error "the version of LabVIEw full development is necessary to correct the errors." A picture of the hierarchy of the VI is attached.
I think I'm missing something in the installation. I'm downloading now the DAQmx. I knowticed that all 7 of my missing Subvi are DAQmx * .vi
I wonder if anyone has come across a simular problem with the synchronization of input and output voltage. I use a box 11 LabView and NI USB-6259. I have been using the DAQ Assistant to configure the input and output channel. In particular, my task is to generate a single rectangular "pulse" as the output voltage to drive a coil and once the pulse went to get a signal from a sensor of magnetic field and get a power spectrum. This means that the order and the time during which the DAQ Assistant is used is extremely important. For example, the output voltage channel must be opened first for 2 seconds. Subsequently, the channel of input voltage must be open for 1 second, in which the sensor signal is obtained and post-processed. Only after these tasks are performed in this order he can can be repeated in a loop until the experiment is over. I don't know how to trigger data acquisition assistants (one for entry) and the other for the voltage output correctly. Y at - it a trick?
See you soon
Thank you that I wired the error strings but the timing issue was unrelated to it. In the DAQ assistant, I simply had to choose the continuous aquistion of the 'samples' methods 'N-switch' for input and output voltage and all works fine now.
I am trying to create a development machine, where we can test the new code without using our physical hardware. I followed this guide to set up a system of simulation. I get to step 3.2 b, but the device does not appear in the DAQ assistant. MAX, the device self test and gites calibrated successfully, and when I open the test panels, I see some sort of signal. I guess that's a default entry simulated since I didn't that device to look for anything? Note that two devices, I am creating the show upward into the devices section and Interfaces, but that, even after running auto calibrate, automatic Calibration date is not yet specified.
When I try to test the device and create a voltage according to the guide, I can't see a device in the creator of data acquisition task.
Steps 1 and 2 of this guide are of course met. Step 3 is not, but this is not surprising because a simulated device is in device in any case manager. Also, I'm not under RT, so step 4 is satisfied.
Someone at - it ideas?
That would be because the PXI-5124 is a digitizer not an analog input device. You must use the NI SCOPE not NOR DAQmx driver
I'm currently dealing with a continuous data using NOR cDAQ-9174 proposed acquisition and recording of analog input signals of a built-in three-measuring probe.
I built a simple vi using DAQ Assistant to acquire data and write to an output .txt - rather than .tdms using Signal Express.
On a day 10 cycle of data acquisition computer was mistakenly turned off - leaving the empty output .txt file. LabView recovered the VI cut and I wonder if there is a way I can access the data that has been saved by the DAQ Assistant which can be saved in temporary files etc..
I have no idea where that might be, since you cannot delve deeper down into the 'levels' of DAQ assistant as you would a sub - vi.
Just as a note aside to apologize my stupidity - I realize that all the data at the end of the writing task is stupid and completely avoidable... but I worked for a date limit.
Thanks in advance for any help you can provide.
The most likely answer is not, unfortunately. It looks like you were a table of data at every point of the construction and then measure he writes at the end. In this case, unless you have explicitly recorded data in a temporary file, it is located right in volatile memory, waiting for you to do something with it.
I realize that this isn't what you want to hear, as it comes to the time of submission of draft / year...
If you post your VI (preferably version LV2012 or below), I can have a look to see if there is anything obvious.
I tried to set up a channel on a module SCXI in 1120 D, as well as the rest of the SCXI system, using the DAQ assistant, but ran into a problem related to the range of the signal.
According to this, http://sine.ni.com/nips/cds/view/p/lang/en/nid/1658, the maximum voltage is located between 10v - 10v, but when I tried to configure a channel with a range of 0v - 10v, it displays an error indicating that the range of the signal was invalid and that the available range is - 0.005v - 0.005v, could someone suggest what I'm missing or what I need to do to set up?
Thank you for your message.
Can I ask that you have the software gain set on the device? The table on page A-5 manual found here your device that the gain settings must be for some input range.
I imagine you have the gain set to 100 while the available range is ±50 mV. For - 10V - 10V, please adjust the gain to 0.5.
Thank you very much
How to create a task in the DAQ Assistant of LabView for one of our modules cDAQ without actually being connected to the cDAQ module?
You can simulate a large number of instrument supported by DAQmx with Measurement & Automation Explorer.
Right-click on the NOR-DAQmx devices and then create new, then simulate and choose from the list of supported devices.
For the cDAQ chassis first then your module.
Excuse me, but I don't have an English version of MAX and so I did not have the correct translation of command...
When your simulated device is configured, ca use you it with the DAQ Assistant in LabVIEW.
Hello. I use "cDAQ-9178" and "NI 9215" and "NEITHER 9402" are added on. "
However, when I run Labview code, "Error-201003" occurs.
Device not available. Possible causes:
Device is no longer present in the system / device is not powered.
Device is turned on, but was temporarily without electricity / device is damaged
(Error appears as the 1st and 2nd figures below).
(Plans of logic is the figure below).
I could be something with the pilot
Check this box:
Error 201003 to the MAX test panel or all by running the DAQ Assistant
In this forum, they speak the same error:
Spontaneous error code 201003 for acquisition of data PCI configuration
For my first application LabVIEW, I'm looking to automate the calibration of thermocouples by measuring their response at different temperatures in a dry well Sizer. I get temperatures of thermocouples six by SCXI 1303/1102/1600 and have six channels put in place in one of my subVIs in the DAQ assistant.
I compare these values to temperature calibrator that I am acquiring by VISA series in an other Subvi. All these values are written in a .csv file.
Can I import these data into the DAQ Assistant to use for calibration? Is there a simpler way to associate with the channel calibration data? Currently, I could manually copy - paste the cells on the worksheet in the calibration sheet, but that seems just silly.
If there is everything that I could provide to help solve the problem, let me know!
It is possible to automate the calibration screws DAQmx at a lower level, in particular the DAQmx channel property node. Using you can specify advance nationwide and the values on the scale for your channel. I recommend you take a look at this example of the community. The property node configuration will be similar to, as illustrated below:
I have a NI USB-6225 DaqMx I used a couple of years. When I started with LabVIEW, I found the Daq Assistant to the best way to measure the voltage with my Daq etc. My company has purchased another DaqMx NI USB-6225 and now I have a big problem: the Daq Assistant in my old Vi does not work with the new data acquisition. I understand why there is this problem, but I do not know how to solve. I found this code on the forum who finds that Daq is connected:
The problem is that Daq Assistant do not have an entry for it, and it gives me an error if I try to run the code with a different device than the original, I used when I created the code.
Is there a way to solve this, so I don't have to convert all the Assistants Daq normal code?
(1) as the old software is related to 'Dev1' you must rename your new device to this alias and skip/rename the old device (and lack).
(2) rewrite your old software does not become is not dependent on the name of the alias for the data acquisition card...
It's your choice!
I was wondering if it is possible or if anyone has any success making a user interface, such as the DAQ Assistant, where you can switch between a graphical output or an array of the recorded signals. I would like to have this feature in my code .VI. From now on, I'm just a waveform graph in my front panel. I'd love to be able to alternate between this waverform table and a table of output values. I have attached a few screenshots to explain what I mean.
Personally, I like the TAB ideas that have been validated, but you can also use property nodes and the visible value.
Hello NOR Developer area,
I am a Novice of LabVIEW and have seen how helpful you all can be, and if I come to ask for your help.
I'm having some trouble with a VI I built that specifies an input voltage, a SCB - 100 connected to a PCI-6031E and converts this tension in a temperature displayed on a waveform table. The goal is to give a constant reading of the temperature and display it in a chart for as long that the VI is running (and to reset the chart the next time the tracks of VI).
The problems I've encountered currently are:
-After a few minutes of the VI running, I get an error message 200279: tried to read samples that are no longer available. The requested sample was already available, but has since been replaced. (to the DAQ Assistant express VI).
-I don't know how to change my chart so that the minimum value X is both during which the VI was launched and have the maximum X value increases with each iteration of the loop. Currently, I have the VI get the time system and contributing to the property node X scale. This worked for the graph of the voltage, but not for the temperature chart
I appreciate those of you who took the time to read my post.
Thank you all for your help.
Ethan A. Klein
SB candidate in Chemistry & Physics
Massachusetts Institute of Technology
Class of 2015
PS I enclose my VI to give you a better understanding of my current situation.
E A Klein wrote:
Thanks for writing.
What property node is talking?
I do not understand that many different data types. How can I go on the treatment of all the data?
(Did you mean I should wire 'blue' data for mathematical functions rather than using the node property tension?)
In fact, one of the nodes property. I mean specifically the tension property node. But in reviewing, I noticed the other nodes in property for the chart. Just set auto-scaling to the X scale and that should take care of two of the nodes property (right click on the graph, X scale-> AutoScale X Scale). I also recommend placing your mathematical functions in a Subvi to make things easier to read. Attached, that's what I think you're after.
I hope that these small tweaks will speed things up enough to avoid your error. If this isn't the case, then we should begin to look at the design of producer/consumer model or take readings at the same time. It might also be worth looking away the DAQ Assistant and DAQmx real screws. But one step at a time.
Maybe you are looking for
Hi all I'm really new with everything regarding controllers or the relay so that any help would be much appreciated. I want to control 5 valves (Web site) with the following characteristics: Rated voltage: 12 or 24 VDC, available voltage fluctuation:
I tried all the ways suggested to correct this error code and nothing works. Do I really need these updates? What other r there suggestions to solve the problem?
original title: Wireless Zero transactional Service not installed My top of Office XP Sp3 Dell doesn't seem to have the Automatic Configuration Wireless installed under the heading 'Services '. How can I install?
Hi all Problem: what we print on A3 from tray 3 paper, printer always says "tray 1 load regular A3! So frustrating! Although all driver settings software set to A3 paper printer must use from tray 3, continually ask printer load Tray 1 with A3 paper.
I fight with my HP ENVY 15-j059nr Quad Edition laptop PC... I've been a fan of HP and Windows for many years. I'm under Win 8.1 and just moved all my old Pavilion Entertainment PC. I spent hours last night download all updates of proposed pilot. It s