Two measures of different thermocouples 1303
I use a SCXI-1303 module and a module SCXI-1102 and SCXI-1000 chassis. The plan is to measure the temperature using thermocouples type T. 1303 modules (I have 3 of them) are completely new, and I pulled out of the box and put them on my 1102. I was testing my thermocouples to ensure they all worked when I noticed a difference in temperature between some of the thermocouples. When I started using AI16 (0-15 was already in use) temperature readings began to appear about 4 degrees below all other temperatures. Most of the thermocouples read 21 c, which is good, but those channels AI16-19 (I'm not plus far only 19), have been reading of 16 C. So I swapped the thermocouples, who had read the right temperature (from AI15) set AI16 and he showed the same offset of 4 degrees, so should not be a thermocouple problem. I put the 1303 on another module of 1102 and I swapped the 1303 modules, I had the same problem each time. The polarization resistance came in 1303 modules, so I have not tested with them. I get the same temperature difference using "integrated" and "constant" for the source of the CJC. I can hardly believe that it's a hardware problem, scxi-1303 1102 both had the same result. So I was hoping there is something else here I'm missing. I looked in the forums and not found anything quite like my problem. I read the manuals but have not encountered anything that would explain this.
Please let me know if you need more information,
Well, I feel stupid. When you use the wizard daq, you must specify the type of thermocouple for each channel. I've had several channel sent to the type of thermocouple wrong without realizing it. Switching the appropriate type, made them of course work.
Heh, sorry for the useless post
Tags: NI Hardware
I'm just curious, why these two tables are different and are they really different in terms of structure, even if they are from the same source? The left is the exit of tdms get properties and the other is a power of arry auto indexed.
No doubt, you are referring to the visual aspect, since the data are the same. One of them uses a different type of indicator of the chain. You probably click right and choose 'create-> Indicator '. When you perform this LabVIEW will use the same style of indicator as what sub-VI/uses. In case you made directly from the PDM function, but the other was created between the autoindex loop tunnel for, in the second he used everything that was default in your preferences.
original title: Microsoft Netmeeting
Can I use NetMeeting to connect two computers in different physical locations but a local call using NetMeeting to dial a number to the other computer by a traditional dial-up 56K modem via telephone lines, (US Robotics ISA 56 K Win Modem kVoice) dialing the telephone number of the other person & their dial-up Modem too? NetMeeting there software to do this, or do we need to use other software, downloaded or already in Windows XP? Can we use NetMeeting on Windows XP Home Edition or XP Professional? We do NOT want to go through the Internet, as our students do not have at this time of money to afford Internet connectivity. . You want to use NetMeeting for tutoring purposes.
Yes, you can use NetMeeting to connect two computers. Follow the steps in the article.
How to turn on the feature Windows NetMeeting Remote Desktop Sharing in Windows XP Service Pack 2 or Microsoft Windows XP Tablet PC Edition 2005
Can I vMotion VM between two hosts with different version? For example, a virtual machine of esxi 6.0 5.5 esxi and esxi esxi 6.0 5.5 vMotion.
Yes, it will avoid you to move.
I can VM vMotion (without shared storage) between two hosts with different vSwitches standard? The vSwitches on these two hosts have different names and number of NIC, but have the same name of portgroup.
Yes, you can in a Storage VMotion to one Esxi to another with no shared storage...
It is important the portgroup bears the same name... This is the reference for the VM...
Why I get two files with different names but the same image when I save in 2015 CC Photoshop?
Try resetting the PS preferences - Preferences in Photoshop
I want to multiply all the lines of the quantinty column in the table of quantity with the relevant lines of the table of prices price column and get the result of multiplying in the third column. What should I use procedure trigerr? OR IS IT POSSIBLE HOW TO DO IT PLEASE HELP :D
Edited by: 994229 2013-03-15 12:44
/* Formatted on 3/15/2013 3:51:08 PM (QP5 v5.185.11230.41888) */ CREATE TABLE mytable1 AS (SELECT 1 id, 5 VALUE FROM DUAL UNION ALL SELECT 2, 7 FROM DUAL UNION ALL SELECT 3, 8 FROM DUAL); CREATE TABLE mytable2 AS (SELECT 1 id, 4 VALUE FROM DUAL UNION ALL SELECT 2, 12 FROM DUAL UNION ALL SELECT 10, 12 FROM DUAL); SELECT id, mytable1.VALUE, mytable2.VALUE, mytable1.VALUE * mytable2.VALUE product FROM mytable1 FULL OUTER JOIN mytable2 USING (id) ORDER BY id;
ID VALUE VALUE_1 PRODUCT 1 5 4 20 2 7 12 84 3 8 10 12
Let assume that I have an analysis with columns: group_id and sales.
On this basis I would like to create two tables: table1 where group_id = 1 and table2 where group_id = 2.
OBIEE is able to do?
You can do it in a single analysis, but you have to have the column with group_id twice.
Bring, group_id_1, group_id_2 (same column), sale
Now in table 1.
bring group_id_1, sales
Then in the selection steps for group_id_1-> select members-> Action choose keep only-> choose the value of the filter 1
This will filter the table 1 with group_id 1
In table2 group_id_2 porter, sales
Then do the same step for group_id_2 with the value 2.
You now have two reports with different filter in the same report.
If you want you can keep the same name column for the two columns group_id, I just kept group_id_1, group_id_2 for easy reference.
In the first Pro CS6, is it possible to use two sequences with different parameters?
Yes, no problem.
You can make as many sequences with different settings as you want.
I need to create a report script to extract data from two years with different months.
For example: 2009: seven to June
2010: July to December
How can I combine two of them in the script for a single report?
Set up your first extraction for 2009: seven to June
Finish with the bang (!)
Then add in the second extraction for 2010: July to December
Finish with the bang (!)
Essentially two separate report of scripts, but if you put them in one, separated by a bang. It will work one after the other, in the same report, issued to the same file script.
This will accomplish what you're asking.
Order line number, amount of command-line, batch Stock number, quantity of booking
ORD_LINE1, 35, LOT1, 10
ORD_LINE1, 35, BATCH2, 17
ORD_LINE1, 35, LOT3, 8
My problem is that the measure of the amount of command line, when totaled in OBIEE gives me 105 (35 * 3) - when it should be 35.
How can I display a total of 35 for amount of command line, although they show 35 for amount of booking?
My underlying SQL query is very sensitive and so would stay away from the analytical functions. Is there something I can do with hierarchies?
It comes to OBIEE 10 g
Edited by: 934462 may 6, 2013 03:09
Edited by: 934462 may 6, 2013 03:15
It seems that your amount depends on only the number of command line while 'reserved' measure depends on many and line number order. So, you have combined measures of two different grains in the same table. Follow these steps:
1 fix the problem at the level of the database of this division in two separate facts and model them as two logical facts connected via a number of command line only one dimension. It's cleaner, and it should run faster.
2. you can copy the amount ordered as a new logical column but not prescribe any rules of aggregation. Align on the presentation layer, and use it as a common command line attribute in your report. Given that this column would be part of the group by, he can sacrifice some performance.
3 the extent of the quantity ordered, go to the tab of aggregation, see "Based on Dimensions", in the new list, choose MAX() as the rule of aggregation for dimension of command lines (create a logical and associated table dummy command line attribute hierarchy if you don't have them) and SUM() as rule of aggregation for the size of the location of stock. It's a little experimental, I doubt how useful it would be in your scenario.
I have a spreadsheet with multiple sheets, I want to add some features to.
At the present time, I have a cell that refers to a cell in a different sheet, so if I changed the value of the cell on sheet 2, it will update on sheet 1. But I had to update the cell directly on sheet 1, it would not be updated on sheet 2.
I want to have it set up so that if I changed the value to 1 or 2 of the worksheet, the cell is updated in both places. AKA, it forces the two cells to be always the same, regardless of the cell is edited to make the change.
Is this possible in number? My language is clear? Any comment is appreciated. Thanks in advance!
The fundamental problem is that a cell of numbers can contain a value or a formula, but not both at the same time. So when you enter a value in a cell, the value you enter will overwrite any formula or a value that was already there. So I don't think that what you want to do is possible by using formulas.
It is possible using AppleScript, but probably isn't worth it.
Can you explain a little more why you want to do this? There may be another way to organized the document, you don't need this.
Here's my question:
I have two different cards installed in a PC. Specifically, PCI-6224 and PCI-6254. I need to acquire 50 analog inputs (real time). Can I use two cards to do? I mean I can synchronize these two cards and acquire simultaneously several channels at the same time?
Yes. You can use a RTSI cable between them. See the examples of the expedition. If you are using LabVIEW, look at several Synch - Analog Input - Cont Acquisition devices.
I have two different cDAQ - 9178 chasises connected to my computer. Each cDAQ-9178 has a NI 9205 module, and each NI 9205 module collects data on its own dedicated experience. The experiences are not always running at the same time, so at some point there may be only a single cDAQ on collecting data instead of two. In addition, every experience is a vi different attached to it for the two cDAQs are not running the same vi.
I would like to know how I could run two separate cDAQ-9178 (which all use another VI) when you are connected to a computer. Thank you for your help.
Well. Use of VI a cDAQ, a cDAQ doesn't "work" a VI.
If each VI uses its own frame, then you should have no problem having 2 VI parallel running or independently each using a cDAQ.
Each cDAQ will be identified in MAX by one different name, as Dev1 and Dev2 to the other. Each VI has just need to refer on behalf of cDAQ for it.
Hi, I'm an application that measures the speed of the wind using crio-9012 and NI 9423 (DIO)
I have the sensor that generates output voltage at different orientations so when he gets shot it generates a type pwm duty cycle 50% wavefrom
When measured with the higher wind speed, it will be generating the signal with greater frequency
Now using labview, I'm trying to see if he can show me the right of the signal as I saw the scope, but it does not show me a good performance
When I rotate the sensor, the light of NI 9423 seems to blink at the right frequency (meaning work) but led I did LabVIEW flashes at a different frequency, not really to work
I don't see how this happens when both LEDs are managed by the same code, the diffrenece is a material is and the other is software
What should I do?
I think that's the problem. Your code is very simple and there is nothing really go. This means that the LED will change very quickly in the resolution of Nova Scotia. The way you look at the façade of the FPGA VI is made using what we call the Interactive Communication of front panel which is not deterministic. This means that as the target FPGA block diagram continues to run, the host updates the values in the FPGA VI front window as often as possible. But, it is not possible to update the rate the FPGA VI running, nor is it possible to transfer this data to the host at that rate there since both are running at different speeds (FPGA your code runs at ns while the UI can be updated by ms). Let me know if you do not understand this.
If I were you, I would test the flashing by placing a kind of counter and count the number of state changes.
Maybe you are looking for
Hey friends For some time when I am on YouTube watching videos, gets terribly slow Safari, the mouse cursor lagging, and scrolling is not responsive. When to leave Safari and reopening of YouTube, it is very good for a short period of time, but safar
HelloAfter Firefox update to v30 Style Editor started adding line breaks in CSS files.It allows to work with the very annoying CSS files. Screenshot: http://S10.postimg.org/5v6kd7cw9/firefox_style_editor_went_wrong.PNG
OK, so I got the Satellite M55-S325 for awhile.But after a while, he started to read errors when trying to log on to my normal account, so it me logged on temporary.After you perform a system restore and go back to my old account, I backed up my work
MEDEL: HP Pavilion 1c39ca g6Hello.A days fea, I have format my hard drive.Before formatting, there is no problem with internet service. (Bote Wifi and cable internet worked well)After formetting, I installed Windows 7 and downloaded all the drivers f
Hey guys,. Just a few quick questions on the development of themes. 1. for the average theme, how to download it if you expect? 2. in the choice of images and funds, you have to buy images of a place to avoid copyright laws (which is just saving Goog